Fixed Luminex panels are now available off-the-shelf and ready to ship, allowing you to make your discoveries sooner.
|Panels||# Analytes||Cat no./web link|
|Human Th9 / Th17 / Th22||17||LKTM009|
|Human Growth Factor||21||LKTM013|
|Human XL Cytokine||45||LKTM014|
Choose from any of the 45 analytes listed below in our fully customizable Human XL Cytokine Panel
Click on the linked analytes to view performance data compared to other suppliers.
|CD40 Ligand||IL-1 beta||IL-33|
|FGF basic/FGF-2||IL-3||MIP-1 alpha/CCL3|
|Flt-3 Ligand||IL-4||MIP-1 beta/CCL4|
|GRO beta/CXCL2||IL-12 p70||RANTES/CCL5|
The XL Cytokine Luminex High-Performance Assay outperformed other suppliers’ assays when tested for linearity of dilution, a common method for assessing assay accuracy. The results for IL-2 are shown below; click here to see more data.
Our assay development is second to none, and we’ve proven this again by testing our new Human XL Cytokine Panel side by side with panels from three other Luminex suppliers. We tested for linearity of dilution, a common method for assessing assay accuracy, and our XL Cytokine Panel outperformed other suppliers' assays using serum, plasma, or both for every analyte shown below.
Accurate assays yield a straight line on the graphs below at or near 100% because the analyte being measured is accurately detected as samples are diluted. If the line deviates far from 100% as samples are diluted, this is an indicator that the assay is not accurately measuring the analyte of interest. For this study, a known concentration of the appropriate recombinant protein was spiked into serum or plasma and assayed by three-fold serial dilutions for each XL Cytokine Luminex Performance Assay and for each supplier Luminex assay.* Sample values were back-calculated based on the standard curve, normalized to the initial sample dilution, and represented as % recovery vs. sample dilution. Remember, a straight line equals an accurate assay!
*When one or more suppliers are absent for an analyte below this is because the supplier's panel did not include that analyte, we were unable to generate data with that kit, or the performance was comparable to the XL Cytokine Panel.
Click on the graphs for a larger image.
Note - the data collected for the 1:27 dilution in plasma with Supplier B's kit fell outside the standard curve range and was not included.